It is highly desirable to be able to optically probe biological activities deep inside live organisms. By employing a spatially confined excitation via a nonlinear transition, multiphoton fluorescence microscopy has become indispensable for imaging scattering samples. However, as the incident laser power drops exponentially with imaging depth due to scattering loss, the out-of-focus fluorescence eventually overwhelms the in-focal signal. The resulting loss of imaging contrast defines a fundamental imaging-depth limit, which cannot be overcome by increasing excitation intensity. Herein we propose to significantly extend this depth limit by multiphoton activation and imaging (MPAI) of photo-activatable fluorophores. The imaging contrast is drastically improved due to the created disparity of bright-dark quantum states in space. We demonstrate this new principle by both analytical theory and experiments on tissue phantoms labeled with synthetic caged fluorescein dye or genetically encodable photoactivatable GFP.
©2012 Optical Society of America
Our ability to study the microscopic world has been revolutionized by advances in optical imaging technology. However, compared to electron microscopy and magnetic resonance imaging, optical microscopy suffers intrinsically from relatively coarse spatial resolution and superficial penetration depth. As one can imagine, resolution and penetration are two fundamentally coupled physical properties. While the diffraction-limited resolution barrier has been broken by a number of super-resolution fluorescence techniques such as STED, (F)PALM and STORM [1–4], the deepest penetration into scattering samples with sub-cellular resolution is achieved currently by multi-photon microscopy. By employing a nonlinear intensity-dependent optical excitation, multi-photon fluorescence is generated primarily at the focal volume where the probability of absorbing two (or more) photons by the same molecule at the same time is the highest, enabling an intrinsic 3D optical sectioning capability [5, 6]. Such a spatially confined excitation scheme thus permits the capture of fluorescence photons emitted and then scattered from the focus by a wide-field non-descanned detector, thereby dramatically increasing the detection sensitivity of scattered signals. This profound feature ultimately leads to a significant imaging depth into scattering samples (more than three times deeper than that of one-photon confocal microscopy) [7, 8].
As many other optical imaging modalities, a fundamental imaging-depth limit still exists for multiphoton fluorescence microscopy when imaging scattering samples labeled with fluorophores in 3D volume [9–13]. Here we take Fig. 1 as an example: the two-photon fluorescence images of fluorescent beads embedded in a turbid 3D sample quickly vanish with the depth when using constant laser power (Fig. 1a). This seemingly shallow cut-off depth is not the ultimate maximum, as imaging can actually be performed much deeper provided that the incident laser power can be elevated accordingly to compensate for the scattering loss (Fig. 1b). However, such a signal-promoting procedure comes at the cost of deteriorating imaging contrast: at a certain depth, the target beads can no longer be identified from the overwhelming background. Formally, the depth where the in-focus signal and the out-of-focus background are equal to each other is defined as the fundamental imaging-depth limit . Note that the conventional optical sectioning picture that multiphoton fluorescence is generated only within the focal volume breaks down here, because the exponential increase of the incident laser power eventually outstrips the power-law fall-off of the excitation efficiency out-of-focus. Obviously, further increasing incident laser power cannot overcome this contrast-rendered imaging-depth limit.
Largely driven by the desire to perform in vivo deep tissue imaging, there have been tremendous efforts in improving the imaging depth of multiphoton microscopy. Several strategies have been explored, such as adaptive optics designed to pre-compensate for the scattering loss [14,15], imaging with longer excitation wavelength , chemical cleaning reagent , and differential aberration imaging . However, relatively little work has been published on exploring novel imaging probes as a way of improving the image contrast or extending the imaging-depth limit.
Herein we present the application of photo-activatable fluorophores (PAFs) to extend the fundamental imaging-depth limit. PAFs are powerful imaging probes for tracking molecular and cellular dynamics with high temporal resolution, and have recently emerged as the key players in super-resolution fluorescence microscopy [2–4,19–22] including nanoscopy with optical sectioning capability . Unlike traditional fluorophores which always remain in their bright states, PAFs permit photo-induced transitions from dark states to bright states. By preferentially inducing the bright states into the focus and dark states into the out-of-focus background, multiphoton activation and imaging (MPAI) of PAFs can significantly enhance the imaging contrast and extend the depth limit of multiphoton microscopy. We demonstrate our proposal by both analytical theory and imaging experiments on tissue phantoms labeled with synthetic caged fluorescein dye or photoactivatable green fluorescent protein. Therefore, by offering the additional on-off molecular quantum states, PAFs seem to be capable of both breaking the diffraction-limited spatial resolution and extending the imaging-depth limit.
2. Results and discussions
2.1 The fundamental imaging-depth limit of two-photon microscopy
Let us first frame the problem of deep two-photon imaging with regular fluorophores. The fundamental imaging-depth limit can be defined as the point where the in-focus signal and the out-of-focus background are equal [9–13]:
We can now analyze the laser intensity distribution within a scattering sample. In a typical scenario where the fluorophores are distributed throughout the sample volume, the number of out-of-focus fluorophores is almost always much larger than that of the in-focus ones . Therefore, at the fundamental imaging-depth limit as defined in Eq. (1), at the focus should be much larger than the out-of-focus counterpart , despite of the scattering loss. Indeed, in the simple condition of homogeneous fluorophore distribution, i.e., , will be identical between the background and the signal, and consequently, the integral of over a subset layers of the out-of-focus volume (e.g., for any z) will be smaller than that over the focus.
2.2 Reducing the background fluorophore concentration alone can improve the depth limit
Although the sample scattering is seemingly the origin of the limited light penetration, the above theoretical framework suggests that the concentration of background fluorophores should play an important role in determining the depth limit. To separate these two effects, we constructed and imaged a set of “two-layer” samples (Fig. 2 ). 0-5% intralipid emulsions were used as they closely mimic the response of biological tissues to near infrared light. In the absence of scattering intralipid, increasing fluorescein concentration in the background does not affect the two-photon imaging quality of the target fluorescent beads. In contrast, when the scattering effects are strong enough (e.g., 2% and 5% intralipid), the two-photon imaging contrast becomes anti-correlated with the background fluorescein concentration. Thus, both strong sample scattering and abundant background fluorophores are necessary conditions for the limited imaging-depth. Sample scattering alone does not create poor imaging contrast. As an important insight, images presented in Fig. 2 suggest that, simply reducing the background dye concentration while maintaining the high intralipid content still allows imaging the target beads with increased contrast.
2.3 Theoretical framework of MPAI using dynamic PAFs
Inspired by the above insight, we envision a way to preferentially keep the background fluorophores in the dark states by using PAFs which have to be activated first by multiphoton process before excited by another multiphoton process. Two different modes of experiments are possible: simultaneous or sequential activation and imaging. When PAFs are being activated and imaged by Ia and Ii (one same activation-imaging laser or two combined lasers, where Ia is the local intensity of the activation laser) simultaneously, in Eq. (1) should be replaced by , the time-dependent concentration of the PAFs in the bright state. Quantitatively, will be the product of and the time-dependent multiphoton activation yield, η (t):
When the timescale for PAFs to diffuse or transport out of the laser focal volume is shorter than the frame acquisition time, the accumulation of the activated fluorophores in the background during the scanning process could be neglected. This dynamic condition is rather common, as exemplified by in vivo blood vessel imaging in which fluorophores are injected into the circulatory systems of small animals . Hence, the signal-to-background ratio for dynamic PAFs can be expressed as
As was analyzed earlier, the laser intensity at the focus is much higher than its out-of-focus counterparts at the imaging-depth limit defined by Eq. (1). This would lead to a much higher dark-to-bright conversion yield at the focus: , under a proper pixel dwell time with a non-saturating activation yield. Consequently, we expect a much improved signal-to-background ratio for PAFs than for regular fluorophores:
It is constructive to note that, after the photoactivation (with a non-saturating activation yield), the number of in-focus activated PAFs become comparable to the total number of the activated ones out-of-focus:Equation (6) clearly indicates that the bright states are preferentially induced at the focus whereas the out-of-focus PAFs mostly remain in dark states. As shown in Fig. 3 , such a remarkable disparity of bright-dark states in space would significantly reduce the background contribution from out-of-focus fluorophores.
2.4 Experimental demonstration of MPAI using a caged organic dye
Caged fluorescein, resorufin and rhodamine are the first developed small organic PAFs. They have been applied to study the assembly of tubulin, hydrodynamic flows and cell lineage during embryo development . Here we use the caged fluorescein, which is commercially available, as a proof-of-principle. Figure 4(a) shows the photo-uncaging reaction of caged fluorescein. The dark state can be activated to the bright state by illuminating with UV light as shown by the absorption and fluorescence spectra and uncaging kinetics in Fig. 4(b). For our two-photon application, laser pulses at 750 nm can uncage the protective groups and subsequently excite the bright state. Therefore only one laser is needed in this simultaneous mode, i.e., We again constructed “two-layer” samples. The target object, a droplet of caged fluorescein solution, was placed on a glass coverslip, and a thick layer made of mixed scattering beads and caged fluorescein was inserted between the target and the objective. Control two-layer samples exhibit same physical parameters except for the use of regular fluorescein in both the target droplets and the out-of-focus background layers.
Figure 4(c)-(f) show the comparison between and in our tissue phantoms labeled with dynamic fluorophores. Using control samples, we first determined the proper concentrations of scattering and dye species in the background layer so that imaging with regular fluorescein is approaching the fundamental imaging-depth limit with (Fig. 4c). We then switched to the corresponding PAFs sample. To accommodate the relatively slow photo-activation of caged fluorescein, a long pixel dwell time of ~1.0 ms was used to obtain enough fluorescence signals. The resulting is found to be about 20 in Fig. 4(d), exhibiting a significant improvement of imaging contrast. Encouraged by this, we further elevated the background dye concentration by a factor of 3 for both control and PAFs samples in Fig. 4(e) and 4(f). As expected, is now beyond what the depth limit permits. In contrast, the corresponding PAFs sample exhibits a remarkable
2.5 Theoretical framework of MPAI using static PAFs
If PAFs are rather static during the entire frame acquisition, we then need to consider the spatio-temporal accumulation of the activated fluorophore in out-of-focus background during the raster scanning. In this scenario, while ηS of the signal remains the same as in Eqs. (2)-(4), the effective activation time for ηB will be longer than the pixel dwell time, τ, by a factor of, f (z), which scales with the z-dependent laser beam area. The result of static PAFs hence becomes:Eq. (4). In the condition of low photo-activation yield, we have the following approximationEq. (1), we have inferred that for all z layers. Consequently, the final activation yield will be higher at the focus than in the background. Therefore, the signal-to-background ratio using static PAFs is still higher in Eq. (8) than that with regular fluorophores in Eq. (1).
2.6 Experimental demonstration of MPAI using a photo-activatable fluorescent protein
PAFs also include several fluorescent proteins that can be genetically encoded and fused to other proteins of interest for in vivo imaging of cells and animals. Photoactivatable green fluorescent protein (pa-GFP) is the first member of this family of optical highlighters . As shown in Fig. 5(a) , pa-GFP can be turned on from its initial dark state by two-photon activation in the 750~850 nm range, and the resulting bright state can be readily imaged in the 900~950 nm range .
We prepared tissue phantoms by embedding the E. coli cells expressing fluorescent protein into 3D agarose gel. The resulting sample is highly scattering due to the densely packed E. coli cells. The PAFs in this sample are considered static, as the cells are stationary within the agarose gel. As demonstrated in Fig. 5(b) and 5(c), while the out-of-focus background is overwhelming for cells expressing regular GFP at a depth of 100 µm, MPAI of cells expressing pa-GFP at the same depth offers a satisfactory image contrast. Thus, we have experimentally demonstrated the ability of MAPI in extending the fundamental imaging-depth limit of two-photon fluorescence microscopy.
3. Methods and materials
3.1 Sample preparation
Fluorescent beads were purchased from Invitrogen (F8765: green/yellow 1.0 µm beads; F8859: green/yellow 4.0 µm beads). Fluorescein (Sigma 46960) and CMNB-Caged Carboxyfluorescein (Invitrogen C-20050) were prepared into 10 mM stock solution in DMSO before diluting to final concentrations. Intralipid (Sigma I141) or non-fluorescent polystyrene beads (Sigma CLB9) were used as the scattering species. “Two-layer” samples in Figs. 2 and 4 were prepared as shown in Figs. 6 and 7 : An imaging spacer with a 120 µm thickness (Sigma S7935) was pasted onto a piece of micro coverglass (VWR 48393-220). Inside the spacer, 5 µL of material serving as the scattering background was added. For the activation-imaging experiments of fluorescein and caged fluorescein (Fig. 4c-f), suspension of non-fluorescent 0.9 µm polystyrene beads (Sigma CLB9) was sonicated and diluted (1:5) into 50% glycerol solution with fluorescein or caged fluorescein at a concentration of 1 mM or 3 mM to serve as the scattering background. The background material was then sealed inside the spacer by another piece of micro coverglass on top, giving a sandwich-like background sample. Imaging targets (fluorescent beads or dye droplets) were placed on top of the glass sandwich before imaging. Glycerol solutions were used to minimize evaporation of solvent during the imaging process. In bacteria imaging experiments, BL21 E. coli strains expressing pa-GFP (Addgene plasmid 11911) and Dronpa-3 (a regular GFP version for control imaging, MBL International Corporation) were harvested by centrifugation, resuspended and embedded inside an imaging spacer using 2% agarose gel. E. coli densities of different samples were calibrated based on OD600 measurement to achieve the consistent scattering effect.
3.2 Fluorescence imaging
All the two-photon fluorescence images were taken on a Leica TCS SP5 MP inverted microscope equipped with a Mai Tai HP laser (690-1040 nm tunable emission) and a HCX PL APO CS 20X DRY microscope objective (N.A. 0.70). A non-descanned epi PMT detector placed directly behind the objective was used for fluorescence collection in combination with a 680/SP emission filter. Images were processed with LAS AF software unless otherwise stated. Images of fluorescent beads (Figs. 1 and 2) were taken under 750 nm with a pixel dwell time of ~2.4 µs. Images of caged-fluorescein-based samples (Fig. 4d and 4f) were taken under 750 nm (48mW) with a pixel dwell time of 977 µs to achieve effective uncaging. Images of E. coli expressing regular GFP (Fig. 5b) were taken under 920 nm (65mW) with a pixel dwell time of 98 µs. Images of E. coli expressing pa-GFP (Fig. 5c) were taken under 920 nm (65mW) after 3 frames of two-photon activation under 830 nm (95 mW) with a pixel dwell time of 98 µs. Note that for tissue phantoms labeled with caged fluorescein or pa-GFP, due to the sample scattering, the actual laser power reached at the deep focal plane is much lower than the measured total power. All images were acquired with 512 by 512 pixels.
To summarize, we have presented the application of PAFs in extending the fundamental imaging-depth limit of multiphoton fluorescence microscopy. This represents a novel strategy compared to the existing approaches that focus on reducing sample scattering. Theoretical framework has been provided to describe both the dynamic and static PAFs during image acquisition. Experimentally, we demonstrated MPAI on scattering tissue phantoms labeled with caged fluorescein (dynamic PAFs scenario) or pa-GFP (static PAFs scenario), which serves as an example of synthetic small-molecule and genetically encoded PAF, respectively.
The practical performance of MPAI can be potentially improved by using PAFs with better designed photophysical properties. First, two-photon photo-activation could be much faster for molecules with larger activation cross sections. For instance, caged coumarin has been shown to exhibit 100 times faster uncaging rate via substrate-assisted photolysis . Second, after being photo-activated, many photochromic fluorophores can also be switched off either thermally or by light [23, 29,30]. Such a switching-off property could re-set PAFs in the entire sample and hence allow repeated imaging on the same x-y plane or navigating through multiple layers.
It does not seem to be a mere coincidence that PAFs can play key roles for both breaking the diffraction-limited spatial resolution of fluorescence microscopy and extending the fundamental imaging-depth limit of multiphoton microscopy. In parallel, it is noteworthy to compare this work to stimulated emission reduced fluorescence (SERF) microscopy which was recently proposed for extending the fundamental imaging-depth limit of two-photon imaging . While SERF microscopy is reminiscent of STED microscopy, the PAFs approach presented here can be considered as being related to PALM in terms of harnessing photo-activation. In essence, the coarse spatial resolution and the superficial penetration depth are both rooted in the difficulty of distinguishing identical molecules in space. By offering the additional disparity of (on-off) molecular quantum states, PAFs render the capability of circumventing both resolution and penetration limits.
Zhixing Chen, Lu Wei and Xinxin Zhu contributed equally to this work. We thank Ya-Ting Kao, Fang Xu, Louis Brus, Rafael Yuste, Nicholas Turro, Virginia Cornish, Darcy Peterka, Christophe Dupre and Miguel Jimenez for helpful discussions. We are grateful to Virginia Cornish for sharing lab equipments and Keith Yeager for assistance on Leica microscope. W.M. acknowledges the startup funds from Columbia University, and grant support from Kavli Institute for Brain Science.
References and links
1. S. W. Hell and J. Wichmann, “Breaking the diffraction resolution limit by stimulated emission: stimulated-emission-depletion fluorescence microscopy,” Opt. Lett. 19(11), 780–782 (1994). [CrossRef] [PubMed]
2. E. Betzig, G. H. Patterson, R. Sougrat, O. W. Lindwasser, S. Olenych, J. S. Bonifacino, M. W. Davidson, J. Lippincott-Schwartz, and H. F. Hess, “Imaging intracellular fluorescent proteins at nanometer resolution,” Science 313(5793), 1642–1645 (2006). [CrossRef] [PubMed]
8. R. Yuste, ed., Imaging: A Laboratory Manual (Cold Spring Harbor Laboratory Press, 2010)
9. P. Theer, M. T. Hasan, and W. Denk, “Two-photon imaging to a depth of 1000 microm in living brains by use of a Ti:Al2O3 regenerative amplifier,” Opt. Lett. 28(12), 1022–1024 (2003). [CrossRef] [PubMed]
12. N. J. Durr, C. T. Weisspfennig, B. A. Holfeld, and A. Ben-Yakar, “Maximum imaging depth of two-photon autofluorescence microscopy in epithelial tissues,” J. Biomed. Opt. 16(2), 026008 (2011). [CrossRef] [PubMed]
15. M. Rueckel, J. A. Mack-Bucher, and W. Denk, “Adaptive wavefront correction in two-photon microscopy using coherence-gated wavefront sensing,” Proc. Natl. Acad. Sci. U.S.A. 103(46), 17137–17142 (2006). [CrossRef] [PubMed]
16. D. Kobat, M. E. Durst, N. Nishimura, A. W. Wong, C. B. Schaffer, and C. Xu, “Deep tissue multiphoton microscopy using longer wavelength excitation,” Opt. Express 17(16), 13354–13364 (2009). [CrossRef] [PubMed]
17. H. Hama, H. Kurokawa, H. Kawano, R. Ando, T. Shimogori, H. Noda, K. Fukami, A. Sakaue-Sawano, and A. Miyawaki, “Scale: a chemical approach for fluorescence imaging and reconstruction of transparent mouse brain,” Nat. Neurosci. 14(11), 1481–1488 (2011). [CrossRef] [PubMed]
20. J. Lippincott-Schwartz and G. H. Patterson, “Photoactivatable fluorescent proteins for diffraction-limited and super-resolution imaging,” Trends Cell Biol. 19(11), 555–565 (2009). [CrossRef] [PubMed]
21. M. Heilemann, P. Dedecker, J. Hofkens, and M. Sauer, “Photoswitches: key molecules for subdiffraction-resolution fluorescence imaging and molecular quantification,” Laser Photon. Rev. 3(1-2), 180–202 (2009). [CrossRef]
22. H. L. Lee, S. J. Lord, S. Iwanaga, K. Zhan, H. Xie, J. C. Williams, H. Wang, G. R. Bowman, E. D. Goley, L. Shapiro, R. J. Twieg, J. Rao, and W. E. Moerner, “Superresolution imaging of targeted proteins in fixed and living cells using photoactivatable organic fluorophores,” J. Am. Chem. Soc. 132(43), 15099–15101 (2010). [CrossRef] [PubMed]
23. J. Fölling, V. Belov, R. Kunetsky, R. Medda, A. Schönle, A. Egner, C. Eggeling, M. Bossi, and S. W. Hell, “Photochromic rhodamines provide nanoscopy with optical sectioning,” Angew. Chem. Int. Ed. Engl. 46(33), 6266–6270 (2007). [CrossRef] [PubMed]
24. D. Kleinfeld, P. P. Mitra, F. Helmchen, and W. Denk, “Fluctuations and stimulus-induced changes in blood flow observed in individual capillaries in layers 2 through 4 of rat neocortex,” Proc. Natl. Acad. Sci. U.S.A. 95(26), 15741–15746 (1998). [CrossRef] [PubMed]
27. M. Schneider, S. Barozzi, I. Testa, M. Faretta, and A. Diaspro, “Two-photon activation and excitation properties of PA-GFP in the 720-920-nm region,” Biophys. J. 89(2), 1346–1352 (2005). [CrossRef] [PubMed]
28. Y. Zhao, Q. Zheng, K. Dakin, K. Xu, M. L. Martinez, and W.-H. Li, “New caged coumarin fluorophores with extraordinary uncaging cross sections suitable for biological imaging applications,” J. Am. Chem. Soc. 126(14), 4653–4663 (2004). [CrossRef] [PubMed]
30. M.-Q. Zhu, G.-F. Zhang, C. Li, M. P. Aldred, E. Chang, R. A. Drezek, and A. D. Q. Li, “Reversible two-photon photoswitching and two-photon imaging of immunofunctionalized nanoparticles Targeted to Cancer Cells,” J. Am. Chem. Soc. 133(2), 365–372 (2011). [CrossRef]
31. L. Wei, Z. Chen, and W. Min, “Stimulated emission reduced fluorescence microscopy: a concept for extending the fundamental depth limit of two-photon fluorescence imaging,” Biomed. Opt. Express 3(6), 1465–1475 (2012). [CrossRef]